FUJI EC-250WR5 Video Gastroscope.
1, The image can not display.
2, After replacing the drive plate, there is still no image in the test. CCD has problems.
3,10% of wires inside the light beam were broken.
4, The insertion tube is ok.
5, Biopsy channel was worn.
6, The wire was worn.
7, Light guide bundle has aging conditions and cracks.
Replacement:CCD; Light Bundle; Biopsy Channe; Water Channel; Light Guide Tube; Wire; Bending Rubber
7 working days.
1, Regularly use a leak detector to measure the leak (the area between inflating and red); If notice that the air leaks, send the item for maintenance in time.
2, It needs special management, regular maintenance, inspection, repair, and record.